The MODIS Science Team Meeting will be held April 29 - May 1, 2014 at the Sheraton Columbia Town Center Hotel in Columbia, MD. Information regarding registration, logistics, and agendas is now available here: https://www.signup4.net/Public/ap.aspx?EID=MODI23E
Terra band 10 EV-based RVS: The EV-based m1 and RVS algorithm currently used for bands 8 & 9 was extended to band 10. A special LUT was delivered to OBPG for testing. The results demonstrated that the LUT produced very similar results to the correction applied by OBPG. It was recommended that the new band 10 approach be implemented in the official Collection 6 LUT. A new LUT was delivered containing this update (220.127.116.11_OC) prior to the start of an OBPG reprocessing. MCST will begin a similar implementation in the forward C6 to eventually synchronize the OBPG and forward C6 LUTs.
December 18, 2013 marked the 14 year anniversary of Terra launch. The MODIS instrument on-board the Terra spacecraft has well surpassed its design lifetime, but continues to acquire valuable science data.
Terra MODIS band 30 detector 7 (production order) had a sudden jump on March 21, 2013 and NEdT started to exceed the specification (0.25) on May 4, 2013. Analysis shows that the NEdT exceeds the specification (0.25) for more than 20% of scans within a granule consistently for a period of two weeks, meeting the criteria for a MODIS TEB detector QA change from “Operational” to “Noisy”. MCST has implemented a QA status change for Terra MODIS band 30 detector 7 from “Operational” to “Noisy” in both the C5 & C6 LUT.
The presentations from the MODIS Science Team Meeting, April-2013 are available at the following location: http://modis.gsfc.nasa.gov/sci_team/meetings/201304/