MODIS IOT Aqua Weekly Report
Mission Operations Days: 2020/298 through 2020/304
October 24, 2020 00:00:00 GMT to October 30, 2020 23:59:59 GMT
Aqua Spacecraft and MODIS Instrument Status:
|PM-1 "Aqua" is in Fine Pointing Normal Mode
FM1 MODIS is in B-Side Nominal Science Mode
|Blackbody||A Off; B On (285K)||Nominal Science Mode|
|Calibration Electronics||A Off; B On||Nominal Science Mode|
|Control Processor||A Off; B On||Nominal Science Mode|
|Nadir Door||Unlatched, Open||Nominal Science Mode|
|Space View Door||Unlatched, Open||Nominal Science Mode|
|Solar Diffuser Door||Unlatched, Closed||Nominal Science Mode|
|FDDI Formatter||A Off; B On||Nominal Science Mode|
|FDDI Port||A Off; B On||Nominal Science Mode|
|FIFO Memory||#1 & #2 Off; #3 & #4 On||Nominal Science Mode|
|Format Processor||A Off; B On||Nominal Science Mode|
|Power Supplies||#1 Off, #2 On||Nominal Science Mode|
|PV VIS FPA||A Off; B On||Nominal Science Mode|
|PV NIR FPA||A Off; B On||Nominal Science Mode|
|PV SMIR FPA||A Off; B On||Nominal Science Mode|
|PV LWIR FPA||A Off; B On||Nominal Science Mode|
|PC LWIR FPA||A Off; B On||Nominal Science Mode|
|RC Outgas Heaters||All 3 (CS, IS, OS) Off||Nominal Science Mode|
|RC LWIR FPA Heater||Off||Nominal Science Mode|
|RC SMIR FPA Heater||On (83K)||Nominal Science Mode|
|Scan Assembly||A Off; B On||Nominal Science Mode|
|SDSM||Off||Nominal Science Mode|
||Nominal Science Mode|
|PS1 Survival Heaters||Enabled||Nominal Science Mode|
|PS2 Survival Heaters||Enabled||Nominal Science Mode|
|Timing Generator||A Off; B On||Nominal Science Mode|
|Flight Software||Rev BJ + 1 set of patches
||Nominal Science Mode|
|Inhibit Ids Set||None||
Nominal Science Mode
This Week's Completed MODIS Activities:
Saturday, October 24, 2020
Sunday, October 25, 2020
Monday, October 26, 2020
Tuesday, October 27, 2020
Lunar Roll Maneuver (-16.5075 degrees)
2020/300 22:44:20 MCL MOD_OA03_FR_DAY_RATE.1
2020/300 23:31:54 MCL MOD_OA03_FR_NIGHT_RATE.2 (01:53 early)
2020/301 00:12:41-00:16:27 MCL MOD_OA08_ROLL_MAN.1
2020/301 00:25:06 MCL MOD_OA03_FR_DAY_RATE.1 (01:53 late)
2020/301 01:12:40 MCL MOD_OA03_FR_NIGHT_RATE.2
Wednesday, October 28, 2020
2020/302 03:00:00 MCL MOD_BB_TEMP_SET_270K_B.2 BB Temp Set to 270K for CP B Coefficients
2020/302 21:00:00 MCL MOD_BB_TEMP_SET_280K_B.2 BB Temp Set to 280K for CP B Coefficients
Thursday, October 29, 2020
2020/303 01:00:00 MCL MOD_BB_TEMP_SET_285K_B.2 BB Temp Set to 285K for CP B Coefficients
2020/303 04:00:00 MCL MOD_BB_TEMP_SET_290K_B.2 BB Temp Set to 290K for CP B Coefficients
2020/303 08:00:00 MCL MOD_BB_TEMP_SET_315K_B.2 BB Temp Set to 315K for CP B Coefficients
2020/303 08:00:30 MCL MOD_BB_DCYCLE_FULL.1 Set blackbody heater duty cycle to FULL
2020/303 12:00:00 MCL MOD_BB_TEMP_SET_270K_B.2 BB Temp Set to 270K for CP B Coefficients
2020/303 12:00:30 MCL MOD_BB_DCYCLE_THIRD.1 Set blackbody heater duty cycle to THIRD
Friday, October 30, 2020
2020/304 08:00:00 MCL MOD_BB_TEMP_SET_285K_B.2 BB Temp Set to 285K for CP B Coefficient
This Week's Scheduled MODIS Activities Not Completed:
This Month's Upcoming MODIS Events:
October’s routine activities include a NTDM on 2020/289, an SDSM Closed on 2020/293, an SDSM Open on 2020/293, and a Blackbody WUCD on 2020/302-305.
Non-routine activities include a Lunar Roll on 2020/301.
Next Week's MODIS Events:
Wednesday, November 04, 2020
2020/309 13:19:56-13:20:12 MCL MOD_SRCA_SETUP_L1_L4_L5_CC.1 SRCA setup activity using lamps 1, 4, and 5, and in constant current feedback mode.
2020/309 13:22:56-14:14:38 MCL MOD_OA22_SRCA_FULL_SPECT_1.2
OA-22: SRCA Full Spectral Calibration Part 1 10W-lamp data collection is from 13:38:50-14:14:37.
2020/309 18:36:05-19:10:37 MCL MOD_OA22_SRCA_FULL_SPECT_2.2
OA-22: SRCA Full Spectral Calibration Part 2 10W-lamp data collection is from 18:36:05-19:10:36.
2020/309 19:55:30-19:55:46 MCL MOD_SRCA_SETUP_L1_L4_L5_CC.1 SRCA setup activity using lamps 1, 4, and 5, and in constant current feedback mode.
2020/309 19:58:30-20:49:31 MCL MOD_OA22_SRCA_FULL_SPECT_3.1
OA-22: SRCA Full Spectral Calibration Part 3 10W-lamp data collection is from 20:14:57-20:49:30.
2020/309 21:52:37-22:30:38 MCL MOD_OA22_SRCA_FULL_SPECT_4.1
OA-22: SRCA Full Spectral Calibration Part 4 10W-lamp data collection is from 21:52:37-22:30:37.
Aqua DMU #151 took place on 2020/296.
Aqua Lunar Roll took place on 2020/301.
General Instrument Comments:
FM1 MODIS is in Science Mode. The SVD and NAD are open, the survival
heaters are enabled, and the Blackbody is heated to 285 K.
MODIS Telemetry Trends:
The FM1 MODIS instrument telemetry is trending nominally.
Non-MODIS Significant Events:
The EMOS mission status webpage (which includes instrument data loss
records) can be found at the following URL. Note that a password is now
required to access the EOS mission links web pages. There are directions
at the URL below for how to obtain a username/password.
Limited Life Item Status:
SRCA 10W Lamp #1: 376.5 of 500 hours
SRCA 10W Lamp #2: 188.1 of 500 hours
SRCA 10W Lamp #3: 205.7 of 500 hours
SRCA 10W Lamp #4: 135.0 of 500 hours
SRCA 1W Lamp #1: 532.1 of 5000 hours
SRCA 1W Lamp #2: 307.1 of 5000 hours
Solar Diffuser Door: 3620 of 3022 Movements
Nadir Aperture Door: 1053 of 1316 Movements
Space View Door: 632 of 1316 Movements