MODIS IOT Aqua Weekly Report
Mission Operations Days: 2020/305 through 2020/311
October 31, 2020 00:00:00 GMT to November 6, 2020 23:59:59 GMT
Aqua Spacecraft and MODIS Instrument Status:
|PM-1 "Aqua" is in Fine Pointing Normal Mode
FM1 MODIS is in B-Side Nominal Science Mode
|Blackbody||A Off; B On (285K)||Nominal Science Mode|
|Calibration Electronics||A Off; B On||Nominal Science Mode|
|Control Processor||A Off; B On||Nominal Science Mode|
|Nadir Door||Unlatched, Open||Nominal Science Mode|
|Space View Door||Unlatched, Open||Nominal Science Mode|
|Solar Diffuser Door||Unlatched, Closed||Nominal Science Mode|
|FDDI Formatter||A Off; B On||Nominal Science Mode|
|FDDI Port||A Off; B On||Nominal Science Mode|
|FIFO Memory||#1 & #2 Off; #3 & #4 On||Nominal Science Mode|
|Format Processor||A Off; B On||Nominal Science Mode|
|Power Supplies||#1 Off, #2 On||Nominal Science Mode|
|PV VIS FPA||A Off; B On||Nominal Science Mode|
|PV NIR FPA||A Off; B On||Nominal Science Mode|
|PV SMIR FPA||A Off; B On||Nominal Science Mode|
|PV LWIR FPA||A Off; B On||Nominal Science Mode|
|PC LWIR FPA||A Off; B On||Nominal Science Mode|
|RC Outgas Heaters||All 3 (CS, IS, OS) Off||Nominal Science Mode|
|RC LWIR FPA Heater||Off||Nominal Science Mode|
|RC SMIR FPA Heater||On (83K)||Nominal Science Mode|
|Scan Assembly||A Off; B On||Nominal Science Mode|
|SDSM||Off||Nominal Science Mode|
||Nominal Science Mode|
|PS1 Survival Heaters||Enabled||Nominal Science Mode|
|PS2 Survival Heaters||Enabled||Nominal Science Mode|
|Timing Generator||A Off; B On||Nominal Science Mode|
|Flight Software||Rev BJ + 1 set of patches
||Nominal Science Mode|
|Inhibit Ids Set||None||
Nominal Science Mode
This Week's Completed MODIS Activities:
Saturday, October 31, 2020
Sunday, November 1, 2020
Monday November 2, 2020
Tuesday, November 3, 2020
Wednesday, November 04, 2020
2020/309 13:19:56-13:20:12 MCL MOD_SRCA_SETUP_L1_L4_L5_CC.1 SRCA setup activity using lamps 1, 4, and 5, and in constant current feedback mode.
2020/309 13:22:56-14:14:38 MCL MOD_OA22_SRCA_FULL_SPECT_1.2
OA-22: SRCA Full Spectral Calibration Part 1 10W-lamp data collection is from 13:38:50-14:14:37.
2020/309 18:36:05-19:10:37 MCL MOD_OA22_SRCA_FULL_SPECT_2.2
OA-22: SRCA Full Spectral Calibration Part 2 10W-lamp data collection is from 18:36:05-19:10:36.
2020/309 19:55:30-19:55:46 MCL MOD_SRCA_SETUP_L1_L4_L5_CC.1 SRCA setup activity using lamps 1, 4, and 5, and in constant current feedback mode.
2020/309 19:58:30-20:49:31 MCL MOD_OA22_SRCA_FULL_SPECT_3.1
OA-22: SRCA Full Spectral Calibration Part 3 10W-lamp data collection is from 20:14:57-20:49:30.
2020/309 21:52:37-22:30:38 MCL MOD_OA22_SRCA_FULL_SPECT_4.1
OA-22: SRCA Full Spectral Calibration Part 4 10W-lamp data collection is from 21:52:37-22:30:37.
Thursday, November 5, 2020
Friday, November 6, 2020
This Week's Scheduled MODIS Activities Not Completed:
This Month's Upcoming MODIS Events:
November’s routine activities include an SRCA Spectral on 2020/309, SDSM Closed on 2020/314, SDSM Open on 2020/314, SDSM Closed on 2020/335, and an SDSM Open on 2020/335.
Non-routine activities include a Lunar Roll on 2020/328 and 2020/330.
Next Week's MODIS Events:
Monday, November 09, 2020
2020/314 21:05:53-21:19:51 MCL MOD_OA16_SDSM_SCREEN.2
OA-16: SD/SDSM Screened - B-Side
2020/314 22:44:46-22:58:44 MCL MOD_OA15_SDSM_OPEN.1
OA67-15: SD/SDSM Open - B-Side
Aqua DMU #152 will take place on 2020/323.
Aqua Lunar Roll will take place on 2020/328 and 2020/330.
General Instrument Comments:
FM1 MODIS is in Science Mode. The SVD and NAD are open, the survival
heaters are enabled, and the Blackbody is heated to 285 K.
MODIS Telemetry Trends:
The FM1 MODIS instrument telemetry is trending nominally.
Non-MODIS Significant Events:
The EMOS mission status webpage (which includes instrument data loss
records) can be found at the following URL. Note that a password is now
required to access the EOS mission links web pages. There are directions
at the URL below for how to obtain a username/password.
Limited Life Item Status:
SRCA 10W Lamp #1: 378.9 of 500 hours
SRCA 10W Lamp #2: 188.1 of 500 hours
SRCA 10W Lamp #3: 205.7 of 500 hours
SRCA 10W Lamp #4: 135.0 of 500 hours
SRCA 1W Lamp #1: 532.1 of 5000 hours
SRCA 1W Lamp #2: 307.1 of 5000 hours
Solar Diffuser Door: 3620 of 3022 Movements
Nadir Aperture Door: 1053 of 1316 Movements
Space View Door: 632 of 1316 Movements