Aqua MODIS band 21 QA
After the recent Aqua BB WUCD on 01/31/2015 it was observed that 3 detectors (2, 3,& 9) for band 21 experienced a b1 jump and NEdT increase. The increased NEdT is greater than the specification and has been consistent for over two weeks. MCST procedure dictates a QA LUT update for these detectors to ‘Noisy’ status. This update will be included in the next regular Aqua LUT delivery. The timestamp for the QA change will be 2015031.2100.